The Synergy HT Multi-detection reader is a robotic-compatible microplate .. and J. Sambrook () Molecular Cloning A Laboratory Manual, Cold Spring. View and Download BioTek Synergy HT operator’s manual online. Multi-Mode Microplate Reader. Synergy HT Laboratory Equipment pdf manual download. Manuals and User Guides for BioTek Synergy HT. We have 2 BioTek Synergy HT manuals available for free PDF download: Operator’s Manual.
|Published (Last):||7 February 2004|
|PDF File Size:||19.53 Mb|
|ePub File Size:||2.44 Mb|
|Price:||Free* [*Free Regsitration Required]|
Running A Dispense Protocol optional Absorbance Plate Test Click to close the Data Reduction dialog. Page 55 Insert the accessories box alongside the buotek module box. To do this, depress the small tab pictured below and separate the connectors.
See criteria in text below. Stock Solution Formulation Results Analysis Sensitivity Test 1.
BIOTEK SYNERGY HT OPERATOR’S MANUAL Pdf Download.
Page 17 Some areas of the dispense module can present pinch hazards when the instrument is operating. If leaks are detected, tighten all fittings and repeat the prime.
Computer Control The Synergy HT is completely computer-controlled; it is not equipped with a keypad or its own user interface. Run a System Test Page 50 – Liquid Test 3 Additional minor cosmetic changes throughout.
If prompted to select a reader, select the Synergy HT and click Select a plate type. If sodium hypochlorite bleach was used, perform Rinse the Fluid Lines on the next page. Page Disconnect synergg heater hh thermistor wires.
Ce Mark EN Ensure that the filters remain in their current locations. If you suspect that a Preventive Maintenance Turn the injector heads counterclockwise and gently pull them out of their sockets. Page 97 H12 position is 1.
Installation Dispense module Top foam cable end cap Dispense module cover Inlet tubes 2 Syringes 2 Outlet tubes 4 Bottom foam end cap with cutouts Accessories Figure 9: A monochromator is used for absorbance measurements. For Use with Liquid Tests see Chapter 5. If you have not already done so: Liquid Test 2 Remove the Shipping Panel Figure 2: The illustration should facilitate system setup with a robotic instrument, such as the BioStack Wipe all sides th the plate carrier.
If the four corner wells are within the accuracy range, the reader is in alignment. Fluorescence Tests Using Methylumbelliferone The last number is the lambda table position number. A daily cleaning regimen is the best way to ensure accurate viotek and a long- life for your instrument and dispense module.
Tip priming trough and priming plate BioTek Biottek, Inc. Canadian Department of Communications Class A Click to run the test. Alternate Test Solutions 1.
BioTek Synergy HT Operator’s Manual
Introduction Package Contents Maanual numbers and package contents are subject to change. Models With Injectors 0. Loosen the two screws and slide the PCB to the right and retighten. Alternate Test Solutions Internal Self-test Errors 0d The outlet tubes are opaque PTFE tubes with threaded fittings on each end that are used to deliver fluid from the syringes to the instrument.
Turn off the incubator. In Gen5, this option is found in a Read step within a Procedure.
Internal plate shaking is also supported. In the product introduction section, added note that Synergy HT basecode software version 2. Page 56 Chapter 2: Unpack and Inspect the Reader Page 34 – 2: Removing The Reader’s Shroud Remove two screws from the top center of the rear panel Remove two side screws When reinstalling the shroud, press down firmly on the top to maintain a good biofek seal while tightening the top screws.
BioTek Synergy HT Manuals
Electromagnetic Interference And Susceptibility Getting Started Guide Only qualified technical personnel should perform troubleshooting and service procedures on internal components.
Put the tubes back in. Remove the Shipping Panel Page 35 Page 36 – 3: